Sierra Leone News, Sierra Express Media, for your trusted place in news!
Advertise HereRobe de mariée princesse
Page added on August 26, 2012
The Radiation Protection Board Secretariat has completed a safety assessment inspection of the diagnostic facility at the Holy Spirit Hospitals, Masuba in Makeni. The inspection was part of the ongoing sensitization programme to create awareness about the use and effect of Radiation. The Radiation Protection Board having had its mandate upgraded by the enactment of the Nuclear Safety and Radiation Protection Act 2012 is poised to create awareness about protection of the public and workers and the environment against the dangers resulting from the use of devices or materials capable of producing ionizing radiation. The inspection is also to ensure that the equipment are working properly to prevent the public from undue exposure. (Photo: Lead Inspector Abdulai Abass Kargbo)
At the Holy Spirit Hospital the lead Inspector from the regulatory body Mr. Abdulai Abass Kargbo on inspection of the x-ray machine, discovered that there were two but only one was registered in the of the regulatory authority. He noted that the x-ray machines were being operated by untrained and unqualified Radiographers and dark room assistants.
He also discovered that there were no radiations warning signs and the door was not lead-lined. He noted that there were no exposure lights and that the waiting area was too close to the X-ray room, meaning that the control area was not properly demarcated.
The Medical Director Holy Spirit Hospital, Dr. Patrick Turay said that they had just relocated the X-ray equipment to the new structure and that they were creating a new environment for the patients and workers. He noted that the Hospital catered for everyone. He requested the Radiation Protection Authority to organize training on basic radiation protection for the radiographers. He also revealed that the new X-ray machine was a donation to the Hospital and effort was being made to regularize the registration of the Machine with the Authority
The Inspection Team, comprised Messrs.’ Abdulai Abass Kargbo, Umar Bun Tejan lead inspector, Ms. Martha Tutu George, and Christopher Wallace – Johnson Public Relations Officer.
Stay with Sierra Express Media, for your trusted place in news!
© 2012, Sierra Express Media. All rights reserved.
Search This Site
LATEST NEWS HEADLINES
ALSO IN THE NEWSJournalists and their information
The cardinal function of a journalist, they say, is firstly to inform, but the emphasis here is on accuracy. The journalist has the sacred responsibility to inform the public accurately, giving due regard to ‘facts’ as obtained from ‘a reliable source’. This is the first social function of a journalist. (Photo: Adeyemi Paul, author, and [...]
MORE STORIESDefending Sierra Leone’s First Lady
HAVE YOUR SAYShare you views: What would you like to see more/less of on Sierra Express Media? Send us your ideas and include your name and the area you are writing from.
Our New Website Has a More Efficient LayoutWe invite you to give us your feedback
MORE NEWS HEADLINES